Performance of 1mm<sup>2</sup> Silicon Photomultipliers
- Created: Thursday, 01 May 2008
A silicon photomultiplier (SPM) is a new type of semiconductor detector that has the potential to replace the photo- multiplier tube (PMT) detector in many applications. In common with a PMT detector, the output of an SPM is an easily detectable current pulse for each detected photon and can be used in both photon counting mode and as an analogue (photocurrent) detector. However, the SPM also has a distinct advantage over PMT detectors. The photon-induced current pulse from a PMT varies greatly from photon to photon, due to the statistics of the PMT multiplication process (excess noise). In contrast, the current pulse from an SPM is identical from photon to photon. This gives the SPM a distinct advantage in photon counting applications as it allows the associated electronics to be greatly simplified. Identical pulses also mean that the SPM can resolve the number of photons in weak optical pulses, so-called photon number resolution. This is critical in a number of applications including linear-optics quantum computing.
Each Geiger-output pulse has a duration of typically 40 — 60ns and contains around a million charge carriers and is usually amplified to a useful level (10 — 100mV) by a simple, fast voltage amplifier. The SPM detector can be used in single photon mode, pulse detection mode, and analogue mode for slowly varying or continuous optical signal above the single photon level.
The main performance parameters of an SPM are its photon detection efficiency (PDE), dynamic range, noise rate, and pulse or gain uniformity.
Photon Detection Efficiency
The PDE is determined by the detection efficiency of the individual photon counting microcells and the geometrical efficiency, or fill factor, and can be written as
where η(λ) is the quantum efficiency of silicon at a given wavelength, ε(V) is the carrier Geiger avalanched initiation probability and F is the SPM fill factor. The quantum efficiency of silicon is 80+% over much of the visible spectrum while the avalanche initiation probability of an electron can be as high as 70%. The quantum efficiency and avalanche initiation probability make up the detection efficiency of the individual photon counting microcells. The fill factor is the ratio of the total active area of all the microcells and the total area of the SPM. The SPM fill factor is determined by design constraints and the size of the microcells. In the SPM design there is a fundamental trade-off between the microcell size, the fill factor and the total number of microcells for a given area. The total number of microcells and the PDE determine the dynamic range of the SPM.