Improved Gas Filling and Sealing of an HC-PCF
- Created: Monday, 01 September 2008
Compact hermetic joint is formed to seal connectorized all-fiber gas reference cell.
An improved packaging approach has been devised for filling a hollow-core photonic-crystal fiber (HC-PCF) with a gas, sealing the HC-PCF to retain the gas, and providing for optical connections and, optionally, a plumbing fitting for changing or augmenting the gas filling. Gas-filled HC-PCFs can be many meters long and have been found to be attractive as relatively compact, lightweight, rugged alternatives to conventional gas-filled glass cells for use as molecular-resonance frequency references for stabilization of lasers in some optical-metrology, lidar, optical-communication, and other advanced applications. Prior approaches to gas filling and sealing of HC-PCFs have involved, variously, omission of any attempt to connectorize the PCF, connectorization inside a vacuum chamber (an awkward and expensive process), or temporary exposure of one end of an HC-PCF to the atmosphere, potentially resulting in contamination of the gas filling. Prior approaches have also involved, variously, fusion splicing of HC-PCFs with other optical fibers or other termination techniques that give rise to Fresnel reflections of about 4 percent, which results in output intensity noise.
The V-groove silicon piece that supports the mechanical splice rests on a support plate. One end of a quartz tube is prepared by contouring it to fit over the V-groove silicon piece and support plate. This end of the tube is put in place to cover the splice region, and an epoxy is used to seal the tube to the V-groove silicon piece, the optical fibers, and the support plate.
The other end of the quartz tube is inserted into a plumbing fitting of a vacuum chamber, which is equipped with valves and connected to a vacuum pump for removal of air from the interior of the HC-PCF, and to a gas cylinder and pressure gauge for filling the interior of the HC-PCF with the desired gas. In a typical application, once the HC-PCF has been filled with the gas, the quartz tube is torch-sealed, forming a relatively compact hermetic junction. Alternatively, if the plumbing fitting includes a valve, it can be left in place to enable re-evacuation and/or refilling of the HC-PCF without the necessity of breaking and remaking the splice.
This work was done by Ilya Poberezhskiy, Patrick Meras, Daniel Chang, and Gary Spiers of Caltech for NASA’s Jet Propulsion Laboratory.
This Brief includes a Technical Support Package (TSP).
Improved Gas Filling and Sealing of an HC-PCF (reference NPO-45193) is currently available for download from the TSP library.
Please Login at the top of the page to download.