Light Collimator and Monitor for a Spectroradiometer
- Friday, 29 August 2008
This system handles the optical input and electronic output of two spectrometers.
A system that comprises optical and electronic subsystems has been developed as an infrastructure for a spectroradiometer that measures time- dependent spectral radiance of the daylight sky, in a narrow field of view (having angular width of the order of 1°) centered on the zenith, in several spectral bands in the wavelength range from 0.3 to 2.2 μm. This system is used in conjunction with two commercially available monolithic spectrometers: a silicon-based one for wavelengths from 0.3 to 1.1 μm and a gallium arsenide-based one for wavelengths from 1.05 to 2.2 μm (see figure). The role of this system is to collect the light from the affected region of the sky, collimate the light, deliver the collimated light to the monolithic spectrometers, and process the electronic outputs of the spectrometers.
When the shutter is open, light enters the monolithic spectrometer, electronic circuits in the spectrometers preprocess the outputs of photodetectors (one photodetector for each wavelength band), and the outputs of the spectrometer electronic circuits for the various wavelength ranges are sent to a data- acquisition-and-control subsystem that is part of the present system. When the shutter is closed, the same process takes place, for the purpose of collecting dark-current readings from the photodetector of each wavelength band.
The data-acquisition-and-control subsystem digitizes the spectrometer outputs and further processes them to generate any or all of a variety of useful output data. Among other things, this subsystem subtracts shutter-closed (dark-current) readings from shutter-open readings to obtain corrected spectral-radiance readings. In addition to alternately opening and closing the shutter and taking dark-current readings during the t2 portions of successive cycles, the system can be made to sample dark currents during longer periods (e.g., a dark period of 5 minutes during each hour) to enable identification of anomalies in this system and/or in the spectrometers.
This work was done by Warren Gore of Ames Research Center.
This invention is owned by NASA, and a patent application has been filed. Inquiries concerning rights for the commercial use of this invention should be addressed to the Ames Technology Partnerships Division at (650) 604-2954. Refer to ARC-15714-1.