Large delays needed for time-domain autocorrelations would be realized photonically.
A compact photonic microwave Fourier spectrum analyzer [a Fourier-transform microwave spectrometer, (FTMWS)] with no moving parts has been proposed for use in remote sensing of weak, natural microwave emissions from the surfaces and atmospheres of planets to enable remote analysis and determination of chemical composition and abundances of critical molecular constituents in space.
The instrument is based on a Bessel beam (light modes with non-zero angular momenta) fiber-optic elements (see figure). It features low power consumption, low mass, and high resolution, without a need for any cryogenics, beyond what is achievable by the current state-of-the-art in space instruments. The instrument can also be used in a wide-band scatterometer mode in active radar systems.
The basic advantage of the proposed instrument is its wide bandwidth along with high resolution, enabling microwave hyperimaging of the planetary atmospheres and surfaces. For example, the analyzer will have similar resolution to the Cassini Titan Radar Mapper operating in the scatterometer mode, and will have at least two orders of magnitude wider bandwidth, compared with the same instrument operating in the radiometer mode. This will allow collecting a hundred times more data during the same observation period.
The analyzer has significant advantages for remote detection of chemical components from space. It uses microwave radiation to record the rotational spectrum of a molecule in the microwave spectral region, i.e., between approximately 6 GHz and 40 GHz. For instance, FTMWS can be used for the accurate remote detection of water vapor as well as for the study of hydrogen isotopic ratio.
The analyzer has a very long, mile-range, maximum delay realized with photonics techniques. This capability is ultimately crucial for achieving a high resolution with the Fourier transform spectrum analyzer. To obtain the 1-MHz spectral resolution necessary for resolving the microwave emission spectrum of water in remote sensing, it would ordinarily be necessary to use microwave delay lines having lengths up to 300 m. Such long microwave delay lines would not be practical. However, the instrument would exploit the fact that compact delay lines can be realized photonically.
It has Fellgett (multiplex) advantage taken from Fourier spectroscopy. A typical microwave spectrum analyzer sequentially measures the microwave power within each of a number of narrow spectral bands. The new instrument would simultaneously measure the time-domain autocorrelations of the microwave emission signal of interest using a number of different delays, then calculate the spectrum of that signal by use of a fast Fourier transform. Hence, the instrument would constantly and simultaneously provide data on all the bands of the spectrum.
It has all the advantages of a static Fourier transform spectrometer. There are no moving parts, which eliminates many potential mechanical problems onboard the spacecraft. The instrument has no need for a reference laser since the detector array samples the interferogram always at the same points. The analyzer obtains full interferogram at once so it is insensitive to the flicker noise or fluctuations of the input signal. This is critical, e.g., for spectroscopy of a constantly changing planetary environment.
As shown in the figure, the incoming microwave signal would be amplified and applied to the input terminal of an electro-optical modulator, which would impress the microwave signal as modulation onto a beam of light. The light would enter a Bessel-beam fiber-optic element, which is essentially an optical waveguide that tapers to a wider aperture at its output end. A Bessel-beam fiber-optic element acts as a highly dispersive radiator horn, roughly equivalent to a bundle of optical fibers of different lengths. The dispersion in a Bessel-beam fiber-optic element is so great as to afford delays ranging from about 10 ps to about 1 μs. In this instrument, the light arriving at each location on the output plane of the dispersive optical element would have a different delay, and so an array of photodiodes would be placed on that output plane to sample signals at various increments of delay. The variously delayed microwave outputs of the photodiodes would be used to obtain the required autocorrelation data.
This work was done by Anatoliy Savchenkov, Andrey Matsko, Dmitry Strekalov, and Lute Maleki of Caltech for NASA’s Jet Propulsion Laboratory.
In accordance with Public Law 96-517, the contractor has elected to retain title to this invention. Inquiries concerning rights for its commercial use should be addressed to:
Innovative Technology Assets Management
Mail Stop 202-233
4800 Oak Grove Drive
Pasadena, CA 91109-8099
Refer to NPO-43992, volume and number of this NASA Tech Briefs issue, and the page number.
This Brief includes a Technical Support Package (TSP).
Compact Microwave Fourier Spectrum Analyzer (reference NPO-43992) is currently available for download from the TSP library.
Please Login at the top of the page to download.