At the heart of Mightex (Pleasonton, CA) multi-channel spectrometers is a
high-throughput flat-field imaging spectrograph. Multiple input fibers, each
representing an independent signal channel, are aligned along the input slit
of the imaging spectrograph. The spectrum of each channel is dispersed by a
high-efficiency diffraction grating and then imaged onto a 2D CCD sensor.
Light from each channel occupies different rows on the CCD sensor. All
channels are exposed simultaneously, then rows associated with each channel
are binned together to produce a spectrum for the channel. Fiber channels are
spaced out properly to essentially eliminate crosstalk between adjacent
channels. The standard CCD camera features a 1/2" 1.3MP Sony ICX205 imager with
a 12-bit ADC. Exposure time (integrating time) can be varied between 50us and
This week's Question: Fox Sports offered virtual-reality streams from last week's U.S. Open, a major golf championship in Oakmont, PA. Sports fans who owned the right devices could watch the golf event on the television while using VR for...
Subscribe today to receive the INSIDER, a FREE e-mail newsletter from NASA Tech Briefs featuring exclusive previews of upcoming articles, late breaking NASA and industry news, hot products and design ideas, links to online resources, and much more.