Energy-Based Tetrahedron Sensor for High-Temperature, High-Pressure Environments
- Monday, 02 January 2012
This sensor is applicable in the mining industry or in acoustic applications where energy-based measurements are required.
An acoustic energy-based probe has been developed that incorporates multiple acoustic sensing elements in order to obtain the acoustic pressure and threedimensional acoustic particle velocity. With these quantities, the user can obtain various energy-based quantities, including acoustic energy density, acoustic intensity, and acoustic impedance.
In this specific development, the probe has been designed to operate in an environment characterized by high temperatures and high pressures as is found in the close vicinity of rocket plumes. Given these capabilities, the probe is designed to be used to investigate the acoustic conditions within the plume of a rocket engine or jet engine to facilitate greater understanding of the noise generation mechanisms in those plumes.
The probe features sensors mounted
inside a solid sphere. The associated
electronics for the probe are contained
within the sphere and the associated
handle for the probe. More importantly,
the design of the probe has desirable
properties that reduce the bias errors
associated with determining the acoustic
pressure and velocity using finite sum
and difference techniques. The diameter
of the probe dictates the lower and
upper operating frequencies for the
probe, where accurate measurements
can be acquired. The current probe
design implements a sphere diameter of
1 in. (2.5 cm), which limits the upper
operating frequency to about 4.5 kHz.
The sensors are operational up to much
higher frequencies, and could be used
to acquire pressure data at higher frequencies,
but the energy-based measurements
are limited to that upper frequency.
Larger or smaller spherical probes
could be designed to go to lower or
higher frequency ranges.
The probe was manufactured using four G.R.A.S 40 BH 1/4" microphones embedded in the 1-in. (2.5-cm) sphere. The pre-amplifiers for the microphones are also embedded in the sphere. These microphones are capable of operation in sound fields up to 190 dB, which make them suitable for the rocket plume environment. The LabVIEW data acquisition system acquires the microphone signals from each of the four probes and estimates the acoustic pressure at the center of the probe as the average of the four measured pressures. The acoustic particle velocity is obtained using finite difference techniques to acquire a velocity estimate between each pair of microphones in the tetrahedron design. These six particle velocity estimates are along different directions and estimate the particle velocity at the center point of that side of the tetrahedron. Thus, the user is required to determine the three orthogonal velocity components from these six estimates made. The advantage of using an energy-based probe is that it allows the user to extract additional information regarding the radiation characteristics of the source being investigated.
This work was done by Kent L. Gee, Scott D. Sommerfeldt, and Jonathan D. Blotter of Brigham Young University for Stennis Space Center. For more information, contact the SSC Chief Technologist Office at (228) 688-1929. SSC-00355