High Speed Cameras for Non-Destructive Testing
- Friday, 01 November 2013
Digital Image Correlation
Over the years, digital image correlation has found widespread popularity in many applications due to its robustness, flexibility, accuracy, and overall ease of use. Although DIC has conventionally been used mostly for quasi-static or moderately dynamic applications, the availability of affordable and reliable high-speed and ultra-high-speed cameras in recent years has led to a dramatic increase in the use of DIC systems for dynamic applications.
Digital Image Correlation utilizes a pair of precisely synchronized high speed cameras to record a random pattern painted onto the side of the object of interest (Figure 1). The recording is then analyzed for shifts in the pixel intensity array subsets on two or more corresponding images. Due to the ease in which DIC can be implemented, it has been widely adopted for micro- and nano-scale mechanical testing applications, and to validate finite element analysis (FEA) models utilized in the development of many products. For example, in the past, in order to validate the computer model, engineers typically placed strain gauges in the predicted high-stress locations, applied a force in the same location and direction as the model, and then analyzed and compared the strain data from the strain gauges to the theoretical data from the FEA model. Typically, the data does not match, and then engineers must determine how to adjust the mechanical testing or the computer model to make sense of the results. Among the many benefits high performance, high-speed cameras provide, it is the ability to precisely synchronize multiple units that has made cameras like the new Fastcam SA-X2 mainstays of the Digital Image Correlation market (Figure 2).
Particle Image Velocimetry (PIV)
Another non-destructive test in which high speed cameras have gained widespread acceptance is the study of particle image velocimetry (PIV) for the analysis of particle movement within a gas or fluid (Figure 3). Here seed particles are very precisely tracked to ascertain the flow within the image sequence. These particles are often very fast moving, and extremely small, so it is critical to keep the time between images as short as possible. Some cameras minimize the time between subsequent frames to the order of a couple of hundred nanoseconds. Then the cameras are precisely synchronized to the laser to trigger it at the end of the odd numbered frames, i.e., frame 1, 3, 5, etc. and at the very beginning of the even frames, frame 2, 4, 6, etc. The PIV software then tracks the seed particles’ motion to provide a very exact map and analysis of the fluid or air flow being studied.
Critical Light Sensitivity
Low cost LED lighting has greatly helped, but it is the sensor design itself that has made the greatest inroads into light sensitivity. Miniature micro lenses have been successfully employed to ensure the maximum possible amount of light is directed onto the light-gathering area of the pixel.
A pixel’s fill factor can roughly be defined as the percentage of the pixel that is used for collecting light, as opposed to the assorted control circuitry also required. As recently as five years ago, a state-of-the-art high speed camera could boast a light sensitivity of around ISO 8,000 for monochrome and 1,600 for color. The reason for this disparity, typically around two to three times between color and monochrome, is the utilization of a color filter array (CFA) over the monochrome sensor from which the color levels for each pixel are calculated. Today, the most light-sensitive high speed camera available delivers an ISO 12232 Ssat (saturation-based sensitivity) qualified light sensitivity of 25,000 for monochrome, 10,000 for color.
Light sensitivity is a very important factor in NDT and other high speed imaging applications. This is why it is essential to carefully evaluate the high speed cameras available, when choosing an imager for a particular application. At first glance, it may not appear that ISO 25,000 represents the most light sensitive camera available, but this is because not all manufacturers accurately report their cameras’ stated light sensitivity. One manufacturer states their cameras are tested to ISO 12232, but then adds a mysterious letter ‘T’ after the stated levels of 43,700 and 3,900 respectively – neither of which are recognized ISO values.
It is also noteworthy that the ‘rule of thumb’, i.e. a color sensor being half to a third as sensitive as its monochrome counterpart, does not apply here. The light-sensitive standard established using ISO 12232 states that monochrome sensors shall be measured using tungsten illumination with an infrared (IR) cut-off filter in place to remove the otherwise overpowering infrared light. When this filter is removed, the amount of light, most of it being unusable infrared light, is significantly increased, hence the outlandish figures noted above. Assuming a usable figure of 43,700 was available (which it is not), ISO speed values dictate that any figure in between 40,000 and 49,999 shall be listed as ISO 40,000.
When evaluating the light sensitive ISO values, always take a critical look at what the stated claims are from all high speed camera manufacturers. If the monochrome value is in excess of 40,000, it does the end user no good as they cannot realize the claimed benefit; it would be like a car manufacturer claiming 1,000 miles per gallon if you were to use their top secret and commercially-unavailable fuel. So make sure when you compare light sensitivity, you compare like-to-like to reveal a true ISO 12232 Ssat value and not something else. Most importantly, insist on a demonstration at your facility so that you can be absolutely certain you are comparing apples to apples.
This article was written by Andrew Bridges, Director, Sales and Marketing, Photron (San Diego, CA). For more information, contact Mr. Bridges at abridges@ photron.com or visit http://info.hotims.com/45609-200.