LVGEMS Time-of-Flight Mass Spectrometry on Satellites
- Created on Monday, 01 April 2013
This technology has applications in plant contaminant monitoring, clinical and medical diagnostics, and homeland security and defense.
NASA’s investigations of the upper atmosphere and ionosphere require measurements of composition of the neutral air and ions. NASA is able to undertake these observations, but the instruments currently in use have their limitations. NASA has extended the scope of its research in the atmosphere and now requires more measurements covering more of the atmosphere. Out of this need, NASA developed multipoint measurements using miniaturized satellites, also called nanosatellites (e.g., CubeSats), that require a new generation of spectrometers that can fit into a 4×4 in. (≈10×10 cm) cross-section in the upgraded satellites. Overall, the new mass spectrometer required for the new depth of atmospheric research must fulfill a new level of low-voltage/low-power requirements, smaller size, and less risk of magnetic contamination.The Low-Voltage Gated Electrostatic Mass Spectrometer (LVGEMS) was developed to fulfill these requirements. The LVGEMS offers a new spectrometer that eliminates magnetic field issues associated with magnetic sector mass spectrometers, reduces power, and is about 1/10 the size of previous instruments. LVGEMS employs the time of flight (TOF) technique in the GEMS mass spectrometer previously developed at Goddard Space Flight Center. However, like any TOF mass spectrometer, GEMS requires a rectangular waveform of large voltage amplitude, exceeding 100 V — that means that the voltage applied to one of the GEMS electrodes has to change from 0 to 100 V in a time of only a few nanoseconds. Such electronic speed requires more power than can be provided in a CubeSat.
In the LVGEMS, the amplitude of the rectangular waveform is reduced to about 1 V, compatible with digital electronics supplies and requiring little power. Thus, the LVGEMS concept makes possible very low power (< 0.5 W) mass spectrometers 1 to 2 in. (≈2.5 to 5 cm) in length; fitting and working well in CubeSats. With less voltage and power, there is also less risk of voltage breakdown at the spectrometer electrodes and less magnetic interference from the supporting electronics. Because of its small size, the LVGEMS can be part of an instrument suite, like the NASA/NRL WINCS (Winds-Ions-Neutral Composition Suite) that provides neutral and ion composition with other instruments providing the neutral wind, ion drift, and temperatures. Perhaps the main advantage of an instrument suite is that instruments like GEMS share electronics and power with other sensors, thus minimizing the power consumed per spectrometer — another enabler for CubeSat missions for ionosphere- thermosphere science.
In an orbiting TOF spectrometer, measurements are difficult because all atoms and molecules species enter at the same velocity; therefore all incident atoms and molecules have the same TOF, making it impossible to differentiate between species. However, the newly developed LVGEMS mass spectrometer provides the ability to accelerate and add the same kinetic energy to all species, and this results in each species having its own unique TOF value for easy identification.
The principal feature of this invention is to enable TOF mass spectrometry in low-Earth-orbit investigations of the thermosphere at the time when new missions require multi-point measurements using large numbers of nanosatellites. LVGEMS offers a solution that requires no magnetic fields, no high-frequency voltages of high amplitudes, uses power less than 0.5 W, and has dimensions measured in a few cm (one inch or less).
This work was done by Federico Herrero of Goddard Space Flight Center. GSC-16255-1