A feed-horn-coupled monolithic array of micromesh bolometers is undergoing development for use in a photometric camera. The array is designed for conducting astrophysical observations in a wavelength band centered at 350 μm. The bolometers are improved versions of previously developed bolometers comprising metalized Si3N4 micromesh radiation absorbers coupled with neutron-transmutation-doped Ge thermistors. Incident radiation heats the absorbers above a base temperature, changing the electrical resistance of each thermistor. In the present array of improved bolometers (see figure), the thermistors are attached to the micromesh absorbers by indium bump bonds and are addressed by use of lithographed, vapor-deposited electrical leads. This architecture reduces the heat capacity and minimizes the thermal conductivity to 1/20 and 1/300, respectively, of earlier versions of these detectors, with consequent improvement in sensitivity and speed of response.
The micromesh bolometers, intended to operate under an optical background set by thermal emission from an ambient-temperature space-borne telescope, are designed such that the random arrival of photons ("photon noise") dominates the noise sources arising from the detector and readout electronics. The micromesh is designed to be a highly thermally and optically efficient absorber with a limiting response time of about 100 μs. The absorber and thermistor heat capacity are minimized in order to give rapid speed of response. Due to the minimization of the absorber volume, the dominant source of heat capacity arises from the thermistor.
The array demonstrates a dark noise-equivalent power of 2.9⋅10-17 W/(Hz)1/2 and a mean heat capacity of 1.3 pJ/K at a detector temperature of 0.390 K from a 0.300 K cold plate. The optical efficiency of the bolometer and feedhorn array, measured by comparing the responses to blackbody calibration sources, lies between 0.4 and 0.6. Photon noise dominates over detector noise arising from phonon, Johnson, and amplifier noise, as measured under the design background conditions. The ratio of total noise to photon noise is found to be 1.21 at an absorbed optical power of 2.4 pW. The array shows high stability with excess noise found to be negligible at frequencies as low as 30 mHz.
This work was done by James Bock and Anthony Turner of Caltech for NASA's Jet Propulsion Laboratory. For further information, access the Technical Support Package (TSP) free on-line at www.techbriefs.com/tsp under the Physical Sciences category.